- Research article
- Open Access
Evaluation of microplate immunocapture method for detection of Vibrio cholerae, Salmonella Typhi and Shigella flexneri from food
BMC Microbiology volume 17, Article number: 189 (2017)
Improved methods with better separation and concentration ability for detection of foodborne pathogens are in constant need. The aim of this study was to evaluate microplate immunocapture (IC) method for detection of Salmonella Typhi, Shigella flexneri and Vibrio cholerae from food samples to provide a better alternative to conventional culture based methods.
The IC method was optimized for incubation time, bacterial concentration, and capture efficiency. 6 h incubation and log 6 CFU/ml cell concentration provided optimal results. The method was shown to be highly specific for the pathogens concerned. Capture efficiency (CE) was around 100% of the target pathogens, whereas CE was either zero or very low for non-target pathogens. The IC method also showed better pathogen detection ability at different concentrations of cells from artificially contaminated food samples in comparison with culture based methods. Performance parameter of the method was also comparable (Detection limit- 25 CFU/25 g; sensitivity 100%; specificity-96.8%; Accuracy-96.7%), even better than culture based methods (Detection limit- 125 CFU/25 g; sensitivity 95.9%; specificity-97%; Accuracy-96.2%).
The IC method poses to be the potential to be used as a method of choice for detection of foodborne pathogens in routine laboratory practice after proper validation.
Foodborne pathogens are a growing concern for human illness, death, and food safety and security . The analysis of foods for pathogen presence is a standard practice for ensuring the safety of food, identifying agents of foodborne illness and determining sources of foodborne outbreaks. Conventionally, the microbiological analysis of food involves culture enrichment followed by isolation on selective media . The initial pre-enrichment of a food sample allows for resuscitation of physiologically stressed microbes and grows all bacteria to detectable levels (>10 CFU/ml), followed by a period of selective enrichment to enable growth of the target organism. From there, the pathogen, if present, is isolated on selective agar, and purification and confirmation occur using morphological, biochemical, or physiological tests . Conventional culture methods, however, are often problematic, in that many are time-consuming and require several days to complete, appropriate selective media are not currently available for all bacterial foodborne pathogens, some bacterial pathogens require specific atmospheric or other growth conditions which may be difficult to simulate in the laboratory and some bacterial pathogens may not be culturable by currently available methods . The presence of high background indigenous microflora and complex matrix in food also limits detection of pathogen from food samples. Again, pathogen often exist in viable but non-culturable states in food which cannot be detected by the conventional culture based method . There is a great need for improved methods for foodborne pathogen detection in food matrices. Concentration and separation of pathogens from the food matrix has been the focus of many studies investigating ways to improve sample assay detection limits and speed time to results.
Immunocapture (IC) to concentrate target pathogen from food samples offers a better alternative to traditional pre-enrichment and enrichment steps for routine analysis in microbiology laboratories [6, 7]. Antibody attached to solid surface can capture bacteria through attachment to cell surface proteins and allow specific isolation of the bacteria from samples with high background flora . IC can be used in combination with culture or molecular methods for isolation and detection of the pathogens. One form of IC, immune-magnetic separation (IMS) (that uses magnetic beads coated with antibody) has been developed against many pathogens such as Salmonella spp. , Listeria monocytogenes , E. coli O157:H7  and Vibrio parahaemolyticus  and is commercially available. IMS has been investigated for the concentration and purification of bacterial pathogens from food samples  and are reportedly more sensitive than comparable conventional culture methods . By using IMS, PCR inhibitors inherent to fecal samples were successfully eliminated , but this approach is limited in routine laboratories of underdeveloped countries due to instrumentation cost.
This study focused on three important foodborne pathogens: Vibrio cholerae, Salmonella Typhi and Shigella flexneri. Many previous reports have showed prevalence of Vibrio cholerae, Salmonella spp. (Salmonella Typhi) and Shigella spp. (Shigella flexneri) in food samples in Bangladesh. Shammi  reported contamination of Vibrio spp., Salmonella spp. and Shigella spp. in chicken, beef, fish and shrimp samples collected from local markets of Dhaka, Bangladesh. Mrityunjoy et al.  also reported prevalence of Vibrio cholerae in different food samples such as chicken, fish, beef and shrimp. Noor et al.  reported Vibrio spp. and Shigella spp. in export oriented shrimp samples of Bangladesh. Fatema et al.  reported prevalence of Salmonella spp. in poultry meat collected from local markets of Bangladesh. Aktar et al.  reported prevalence of Salmonella spp. in meat and dry fish samples in Bangladesh. Prevalence of Vibrio cholerae, Salmonella spp. and Shigella spp. in different types of food samples of Bangladesh were reported also by many other researchers [21,22,23,24].
This study evaluates microplate IC (wells of 96 well polystyrene plate coated with antibody) as a better alternative to conventional methods for detection of foodborne pathogens.
To analyze the capture efficiency of the microplate IC method, bacterial cell was added to the wells and incubated for varying periods (1-24 h). Capture efficiency was increased from 1 h (about 40%) to 6 h (about 80%) gradually, after that it was increased slowly till 24 h (Fig 1). This trend was similar for all three pathogens. Results indicate that 6 h incubation for IC could be considered sufficient to detect the pathogens. To determine optimum inoculum cell density, different concentration of cells was added to IC wells for 6 h and capture efficiency (CE) was determined. Results showed that CE is satisfactory (~80%) up to log6 cfu/ml cell concentration, but at higher cell concentration CE declines rapidly (Fig 2). This may be due to limited space and bound antibody in the wells of 96 well plate. On the basis of the results of Figs. 1 and 2, the optimized cell density and reaction time was selected to be log6 CFU/ml and 6 h respectively.
To determine the specificity of microplate IC, detection of other related pathogens has been tested. The method showed considerable specificity to detect Vibrio cholerae, Shigella flexneri and Salmonella Typhi (CE was around 100%). CE was zero or very low in case of other bacteria tested (Fig. 3). One of the potential disadvantages of this method is the chance of cross reaction with closely related species, as it occurs in this experiment. Experiments with Salmonella Enteritidis, Salmonella Typhimurium and Vibrio parahaemolyticus showed some CE may be due to cross reaction, though the CE is not significant (P < 0.05).
For the detection of the target pathogens in artificially contaminated samples, both IC-culture and IC-PCR showed better sensitivity than the traditional culture method and direct PCR (Table 1). For example, when food samples were inoculated with 101 CFU/25 g bacteria, the IC-culture method can detect S. Typhi from 13.3% minced beef samples, S. flexneri from 16.7% minced chicken samples, V. cholerae from 20% minced fish samples and all the three pathogens simultaneously from 26.7% minced shrimp samples. The IC-PCR method can detect S. Typhi from 20% minced beef samples, S. flexneri from 16.7% minced chicken samples, V. cholerae from 20% minced fish samples and all the three pathogens simultaneously from 33.3% minced shrimp samples. In contrast, the traditional culture method was unable to detect any pathogen in these samples and direct PCR can detect S. Typhi from 20% minced beef samples, S. flexneri from 11.1% minced chicken samples, V. cholerae from 20% minced fish samples and all the three pathogens simultaneously from 20% minced shrimp samples. When food samples were spiked with 102 CFU/25 g bacteria, IC-culture and IC-PCR method can detect the pathogens from more than 30% samples, whereas culture method’s detection range was very low (6.7-13.3%) (Table 1). At higher concentration of bacteria in food (~105 CFU/25 g), the detection ability of the IC and culture method were close (100% in the case of IC-culture and IC-PCR and around 90% in the case of culture method) (Table 1). At all bacteria concentration, IC-culture and IC-PCR method showed better detection performance than direct PCR. The detection ability of the IC (both IC-culture and IC-PCR) method was found to be independent of the origin of the samples. Comparing all the food samples in total, both IC-culture and IC-PCR (can detect pathogens in around 60% of the samples) showed a better sensitivity than the culture method (detect pathogens in around 40% of the samples) and direct PCR (detect pathogens in around 55% of the samples) (Table 1).
Performance evaluation of the microplate IC method with the conventional culture based method showed comparable performance of the method for detection of the pathogens (Table 2). Detection limit of the IC-PCR method (25 CFU/25 g) and IC-culture (50 CFU/25 g) was lower than the culture method (125 CFU/25 g). Detection limit of IC-culture was equal to direct PCR (50 CFU/25 g), but the detection limit of IC-PCR was lower than that of direct PCR. Sensitivity of IC-PCR (100%) and IC-culture (97.3%) was slightly higher than culture method (95.9%), and specificity and accuracy of IC-PCR (97% & 96.2%) and IC-culture (96.1% & 95.3%) was slightly lower than culture method (96.8% & 95.3%) (Table 2). Direct PCR (detection limit- 50 CFU/25 g; sensitivity 98.2%; specificity 95.4%; accuracy 96.1%) has comparable performance to IC-culture (detection limit- 50 CFU/25 g; sensitivity 97.3%; specificity 96.1%; accuracy 95.3%) but the IC-PCR showed better performance (detection limit- 25 CFU/25 g; sensitivity 100%; specificity 96.8%; accuracy 96.7%) than direct PCR.
Improvements in the microbiological safety of foods have been largely driven by public demand in response to disease outbreaks . The ability to analyze food products for the presence of pathogenic bacteria is essential for verifying the safety of foods, identifying agents of foodborne illness and determining sources of foodborne outbreaks. Conventionally, the microbiological analysis of food involves culture enrichment followed by isolation on selective media . Confirmation of presumptive isolates is generally through biochemical characterization and/or serology. Such methods suffer from a number of drawbacks. In this study, we assessed the suitability of antibody coated 96 well microplates for the detection of three foodborne pathogen, namely Vibrio cholerae, Salmonella Typhi, and Shigella flexneri.
To optimize the microplate IC method for detection of the pathogens, the effect of incubation time on the capture efficiency of the method was determined and results showed that 6 h incubation is sufficient for optimum capture of pathogens from food samples (Fig. 1). The effect of bacterial density on the IC method was also determined and result showed that CE is satisfactory up to log 6 CFU/ml of the target pathogens (Fig. 2). Specificity of the IC method was evaluated and the method showed significant specificity for detection of the target pathogens, Salmonella Typhi, Shigella flexneri, and Vibrio cholerae. The method can discriminate the pathogens from closely related species, too (Fig. 3). Suitability of the IC method for detection of pathogens in food samples was tested with artificially contaminated food samples. Food samples were spiked with different concentration of pathogens, and the IC method and culture method were employed to detect the pathogens. Results of the comparison of IC and culture method have been summarized in Table 1. In the case of minced beef, S. Typhi was detected in 40% of samples by conventional culture based method, whereas the pathogen was detected in 56% sample by direct PCR and IC-Culture and IC-PCR method detected S. Typhi in 62.7% of the samples. In minced chicken, S. flexneri was detected in 38.8% of samples by culture method, in 58.8% of samples by direct PCR, in 60% of samples by IC-Culture, and in 67.8% of samples by IC-PCR. In minced fish samples, V. cholerae was detected in 42.7% of samples by culture method, in 58.7% of samples by direct PCR, in 60% of samples by IC-Culture, and in 65.3% of samples by IC-PCR. In the case of minced shrimp, three pathogens (S. Typhi, S. flexneri, V. cholerae) were detected simultaneously and results showed that the pathogens were detected in 40% of samples by conventional culture based method and in 52% sample by direct PCR and in 65.3% of samples by IC-Culture, whereas IC-PCR method detected the pathogens in 69.3% of the samples. Finally, comparison of performance parameters of IC and culture method showed that the IC method offers better performance than the culture method for detection of pathogens from food in terms of the detection limit, specificity, sensitivity, and accuracy (Table 2). The detection limit of IC-PCR (25 CFU/25 g) was shown to be lower than that of the culture method (125 CFU/25 g). Other performance parameters of IC-PCR (sensitivity- 100%, specificity- 96.8%, accuracy- 96.7%) was also improved while compared with culture method (sensitivity- 95.9%, specificity- 97%, accuracy- 96.2%) (Table 2).
Different variations of IC have been developed and reported earlier, most of which are based on immune-magnetic separation (IMS). Xiong et al.  reported an IMS method that can capture E. coli O157:H7 from food samples with high capture efficiency (>98%) and specificity. Yang  reported a di-electrophoresis assisted immune-capture for detection of foodborne pathogens with moderate capture efficiency (~60%). Wang et al.  reported an IMS-PCR method for detection of Alicyclobacillus acidoterrestris in apple juice has better sensitivity, specificity, and accuracy (90.9%, 97.0% and 96.2%, respectively). Conceicao et al.  reported an IMS-Plating method with antibody coated polystyrene microspheres for detection of Salmonella sp. in chicken cut with high sensitivity (100%) and specificity (94%).
The microplate immunocapture (MIC) method developed in this study offers a better alternative to the previously reported, IMS methods in terms of ease of operation as it obviates the use of magnetic beads and separation systems. The MIC method also showed competitive performance (sensitivity, specificity and accuracy) with the above mentioned IMS methods. Future research is needed to resolve some shortcomings of this method, such as the effect of high microbial background in food samples. More advanced methods such as whole genome sequencing may provide a more accurate and precise identification scheme for detection of bacteria from food, but in resource-limited laboratories, simpler PCR based method is still the preferable method.
This study showed that antibody coated microplate can be used for detection of foodborne pathogens from a consortium of non-target organisms with high specificity and sensitivity. This method provides an alternative to the conventional culture based microbiological detection method. The microplate IC method is simple, expensive equipment is not needed and can be modified to detect multiple pathogens simultaneously. From preliminary screening, the method showed promise, but further detailed studies are needed to validate the method for use as a routine method for food laboratories.
Chemicals and reagents
All the media used were purchased from HiMedia (India). Antibodies used were as follows- V. cholerae polyvalent antiserum (serovar Inaba & Ogawa); Salmonella polyvalent O antiserum; Shigella flexneri polyvalent antibody (Denka Seiken, Tokyo, Japan).
Vibrio cholerae ATCC-17802, Salmonella Typhi ATCC-65154, and Shigella flexneri ATCC-12022 were taken from culture collection pool of industrial microbiology laboratory, IFST, BCSIR, Dhaka. The list of other strains used in this study is provided in Table 3.
Preparation of antibody coated microplate
Commercial antibody was diluted 1 to 100 in sterile PBS (10 μl antibody mixed with 990 μl of PBS; working concentration- 10 μg/ml) and 200 μl was added to each well of a polystyrene microplate (Nunc™ Microwell™ 96 μ-well plate, Thermo Scientific, USA) and incubated at 40 C overnight covered with aluminum foil. On the next day, antibody solution was discarded and the microplate was washed 3 times with sterile PBS. For blocking, 200 μl of blocking reagent (1% BSA dissolved in 1X PBS) was added to each well and the plate was incubated for 2 h at 37 °C. After blocking, reagent discarded the wells were washed 3 times with PBS (+10% Tween) solution.
Capture efficiency (CE)
Capture efficiency was defined as the percentage fraction of the total bacteria captured by the antibodies in the well and was calculated using a method based on the cells unbound in the well or left in the supernatant. Following equation was used for CE calculation-.
CE (%) = (1-B/A)×100%.
Where A is the total number of cells present in the sample (CFU/ml) and B is the number of cells unbound in the well (CFU/ml, in the supernatant and washed solution). Number of unbound cells in the well were counted based on optical density.
For selection of optimum cell density for inoculation and incubation time, capture efficiency was determined with different inoculum cell density (1, 2, 4, 6, 8, and 10 Log CFU/ml) and with capture times ranging from 1 to 24 h (cell density of log 6 CFU/ml).
To test the specificity as well as the validity of the IC method, and to ensure that the method does not amplify closely related pathogens, a number of closely related and non-target organisms were included in the study (Table 3). The bacteria were cultured in LB broth at 370 C for 12 h, and serially diluted to approximately log6 CFU/ml in PBS containing 1% BSA. Immunocapture was performed and capture efficiency was determined as described above.
Immunocapture of bacteria
Reference cultures of bacteria were grown in LB broth overnight (condition: temperature- 370 C, shaking- 120 rpm) and enumerated (by drop plate method using nutrient agar media) and cell suspension of different concentrations was prepared. Two hundred microliters of cell suspension was added to respective wells and the plate was incubated at 370 C under shaking condition (100 rpm) for 6 h. A negative control well was used to check cross-contamination. After incubation, the medium was discarded and the wells were washed with sterile PBS and fresh LB broth was added. Bacteria from the well were scraped with sterile loop and streaked onto selective agar plate for respective bacteria (TCBS for Vibrio cholerae; SS agar for Salmonella and Shigella) and the plates were incubated at 370 C overnight. Colonies developed on the agar were sub-cultured onto LB agar and identification was done either by culture method or PCR.
Isolation of bacteria from food samples
For spiking of food samples, 250 g of coarsely ground portion of each food type (beef/chicken/fish/shrimp) was used. Spiking was done to obtain different concentration of target bacteria (S. Typhi in beef, S. flexneri in chicken, V. cholerae in fish and S. Typhi, S. flexneri, V. cholerae in shrimp). The Pathogen was grown in LB broth overnight and enumerated and diluted in 10 ml sterile PBS to different concentrations (102 CFU, 103 CFU, 104 CFU, 105 CFU, 106 CFU). Ten 10 ml bacterial suspension was mixed well with 250 g food portion under aseptic condition. Twenty-five grams of representative sample was homogenized in 225 ml of 1% peptone water by stomaching (25 stroke in a sterile bag) (BagMixer®400 W, InterScience, USA) and a part (1 ml) of the stomached sample was used for conventional isolation method with pre-enrichment and enrichment steps. From the enriched samples, 1 ml was used for DNA extraction for detection of pathogen by PCR (hereby referred as direct PCR) and 1 ml was spreaded onto selective media for cultural detection of pathogens. Another portion (1 ml) was used for isolation by IC (both IC-culture & IC-PCR) method (Fig. 4). After IC (as described above), DNA was extracted from microplate well for IC-PCR. For IC-culture, the well was washed with the original media vigorously and the media were inoculated onto selective media by spread plating method. Four types of food samples were used- minced beef (n = 5), minced chicken (n = 6), minced fish (n = 5), minced shrimp (n = 5). In case of spiked shrimp sample, all three antibodies were tested together. In other samples, a single antibody was used.
Identification of bacteria
For detection of Salmonella spp., food sample homogenate in buffered peptone water was pre-enriched at 350 C for 24 h and 1 ml of that was transferred to 10 mls tetrathionate broth and incubated for 24 h at 350 C. The enrichment broth was then plated onto selective media (SS agar and XLD agar) by streaking and the plates were incubated at 350 C for 24 h. In the case of IC-culture, media from each well was spread onto selective media (SS agar and XLD agar) and incubated at 350 C for 24 h. After 24 h, typical Salmonella colonies (Black centered colony on SS agar and pink colonies on XLD agar) were selected and identified by biochemical test as described in ISO 6579-1:2002 .
For detection of Vibrio cholerae, food sample homogenate in buffered peptone water was enriched at 350 C for 24 h. The enrichment broth was then plated onto selective media (TCBS agar) by streaking, and the plates were incubated at 350 C for 24 h. In the case of IC-culture, media from each well was spread onto selective media (TCBS agar) and incubated at 350 C for 24 h. After 24 h, typical Vibrio cholerae colonies (large, smooth, yellow and slightly flattened with opaque centers and translucent peripheries) were selected and identified by biochemical test as described in ISO/TS 21872-1:2007 .
For detection of Shigella spp., food sample homogenate in buffered peptone water was pre-enriched at 350 C for 24 h and 1 ml of that was transferred to 10 mls Shigella- broth and incubated for 24 h at 350 C. The enrichment broth was then plated onto selective media (SS agar and XLD agar) by streaking, and the plates were incubated at 350 C for 24 h. In the case of IC-culture, media from each well was spread onto selective media (SS agar and XLD agar) and incubated at 350 C for 24 h. After 24 h, typical Shigella spp. colonies (olorless colony on SS agar and red colonies on XLD agar) were selected and identified by biochemical test as described in ISO 21567:2004 .
Genomic DNA from pure culture of bacteria and from enriched samples was extracted by Accuprep® genomic DNA extraction kit (Cat. No.: K-3032) by the supplied procedure. For extraction of DNA from microplate wells, 200 μl of lysis buffer containing proteinase K was added to each well and the plate was incubated at 700 C for 30 min for lysis. The lysate mixture was transferred to an Eppendorf tube, and 200 μl 100% ethanol was added and mixed prior to centrifugation. After centrifugation, the pellet was washed with 70% ethanol, dried for 15 mins, and resuspended in miliQ water. DNA concentration was measured by Nanodrop spectrophotometer.
Detection of pathogen by PCR
Primers used for PCR are listed in Table 4. PCR reaction was performed according to the references provided [21,22,23]. PCR master mix (30 μl) was prepared as follows: sterile de-ionized water-23.2 μl, 10X PCR buffer with Mg-3 μl, 10 mM dNTP mixture- 0.5 μl, forward primer (10 mM)- 1 μl, reverse primer (10 mM)- 1 μl, Taq DNA polymerase- 0.3 μl and template DNA- 1 μl. Each PCR tube containing the appropriate mixtures was heated at 95 °C for 3 min in the thermal cycler (BioRad, USA) to ensure the complete denaturation of DNA templates. The PCR was then continued with the following programs: denaturation for 1 min at 94 °C, annealing for 1 min at 55 °C, and extension at 72 °C, for 1 min. Thirty five (35) cycles of these segments were repeated with a final extension of 10 min at 72 °C. PCR tubes were then stored at −20 °C until further analysis. PCR products were gel electrophoresed in 1% agarose gel for visualization of target band, 470 bp for sefA , 588 bp for ompW  and 423 bp for ipaH .
Sensitivity was determined as TP/(TP + FN) × 100, where TP stands by the number of true positive results and FN is the number of false negative results. Specificity was determined as TN/(TN + FP) × 100, where TN stands for the number of true negative results and FP is the number of false positive results. Accuracy was determined as the number of correct results divided by the number of all returned results.
The study was replicated three times. All the results are expressed as mean ± standard error (SE). The values were compared using the analysis of variance followed by post-hoc multiple comparisons using the Least Significant Difference (LSD) test (v9.1; SAS for Windows, Cary, NC) and differences were considered significant when P < 0.05.
American Type Culture Collection
Bovine serum albumin
Colony forming unit
Immunocapture-Polymerase Chain Reaction
International Organization for Standardization
Phosphate buffered saline
Thiosulfate citrate bile salt sucrose
Xylose Lysine Deoxycholate; Min- Minute
Mandal PK, Biswas AK, Choi K, Pal UK. Methods for rapid detection of foodborne pathogens: an overview. Am J Food Technol. 2011;6(2):87–102.
Fu S, Guo S, Gao S, Shen Z. Recent advances in the methods for rapid detection of foodborne pathogens. Asian J Agri Food Sci. 2014;2(5):403–8.
Zhao X, Lin CW, Wang J, Oh DH. Advances in rapid detection methods for foodborne pathogens. J Microbiol Biotechnol. 2014;24(3):297–312.
Taban BM, Ben U, Aytac SA. Rapid detection of Salmonella in milk by combined immunomagentic separation-polymerase chain reaction assay. J Dairy Sci. 2009;92:2382–8.
Hsu C-Y, Hsu B-M, Chang T-Y, Hsu T-K, Shen S-M, Chiu Y-C, Wang H-J, Ji W-T, Fan C-W, Chen J-L. Evaluation of Immunomagnetic separation for the detection of Salmonella in surface waters by polymerase chain reaction. Int J Environ Res Public Health. 2014;11:9811–21.
Arbault P, Desroche N, Larose D. Methods for immunocapture and concentration of bacteria in a sample. Patent no- WO2014140381A1. 2014.
Arbault P, Delphine L, Desroche N, Guzzo J. Microplate Immunocapture (IMC): A New Solution for the Isolation/Concentration of Escherichia coli O157:H7 from Food Samples. Presented at the 2014 meeting for the International Association of Food Protection (IAFP) https://iafp.confex.com/iafp/2014/webprogram/Paper6628.html
Wang Z, Wang J, Yue T, Yuan Y, Cai R, et al. Immunomagnetic separation combined with polymerase chain reaction for the detection of Alicyclobacillus acidoterrestris in apple juice. PLoS One. 2013;8(12):e82376.
Jordan D, Vancov T, Chowdhury A, Andersen LM, Jury K, Stevenson AE, Morris SG. The relationship between concentration of a dual marker strain of Salmonella Typhimurium in bovine faeces and its probability of detection by immunomagnetic separation and culture. J Appl Microbiol. 2004;97:1054–62.
Kaclikova E, Kuchta H, Kay H, Gray D. Separation of Listeria from cheese and enrichment media using antibody-coated microbeads and centrifugation. J Microbiol Methods. 2001;46:63–7.
Chapman PA, Ashton R. An evaluation of rapid methods for detecting Escherichia coli O157 on beef carcasses. Int J Food Microbiol. 2003;87:279–85.
Datta S, Janes ME, Simonson JG. Immunomagentic separation and coagglutination of Vibrio parahaemolyticus with anti-flagellar protein monoclonal antibody. Clin Vaccine Immunol. 2008;15(10):1541–6.
Lynch MJB, Leon-Velarde CG, McEwen S, Odumeru JA. Evaluation of an automated immunomagnetic separation method for the rapid detection of Salmonella species in poultry environmental samples. J Microbiol Methods. 2004;58:285–8.
Hsih HY, Tsen HY. Combination of immunomagnetic separation and polymerase chain reaction for the simultaneous detection of Listeria monocytogenes and Salmonella spp. in food samples. J Food Prot. 2001;64:1744–50.
Islam MA, Heuvelink AE, Talukder KA, Zwietering MH, De Boer E. Evaluation of immunomagentic separation and PCR for the detection of Escherichia coli O157 in animal feces and meats. J Food Prot. 2006;69(12):2865–9.
Shammi T. Detection of Vibrio spp., Salmonella spp., and Shigella spp. among the frozen food samples employing enrichment culture technique. Stamford J Microbiol. 2015;5(1):26–9.
Mrityunjoy A, Kaniz F, Fahmida J, Shanzida JS, Aftab MU, Rashed N. Prevalence of Vibrio cholerae in different food samples in the city of Dhaka. Bangladesh Intl Food Res J. 2013;20(2):1017–22.
Noor R, Hasan MF, Rahman MM. Molecular characterization of the virulent microorganisms along with their drug-resistance traits associated with the export quality frozen shrimps in Bangladesh. SpringerPlus. 2014;3:469.
Fatema K, Rahman M, Datta S, Magnet MH. Comparative analysis of multi-drug resistance pattern of Salmonella sp. isolated from chicken feces and poultry meat in Dhaka city of Bangladesh. IOSR J Pharma Biol Sci. 2014;9(2):147–54.
Aktar N, Bilkis R, Ilias M. Isolation and identification of Salmonella sp. from different food. Intl J Biosci. 2016;8(2):16–24.
Islam S, Nasrin N, Rizwan F, Nahar L, Bhowmik A, Esha SA, Talukder KA, Akter M, Roy A, Ahmed M. Microbial contamination of street vended foods from a university campus in Bangladesh. Southeast Asian J Trop Med Public Health. 2015;46(3):480–5.
Islam A, Mazumdar RM, Fakruddin M, Islam S, Nipa MN, Iqbal A, Bhuiyan HR. Multiple antibiotic resistant bacteria on fruits from different markets of Chittagong city in Bangladesh. Bangladesh Res Pub J. 2010;4(4):332–40.
Nipa MN, Mazumdar RM, Hasan MM, Fakruddin M, Islam S, Bhuiyan HR, Iqbal A. Prevalence of multi drug resistant bacteria on raw salad vegetables sold in major Markets of Chittagong City, Bangladesh. Middle-East J Sci Res. 2011;10(1):70–7.
Fakruddin M, Rahman MM, Ahmed MM, Hoque MM. Antimicrobial resistance and virulence factors of enterobacteriaceae isolated from food samples of Bangladesh. Intl J Microbiol Immunol Res. 2014;3(1):12–8.
Newell DG, Koopmans M, Verhoef L, Duizer E, Aidara-Kane A, Sprong H, Opsteegh M, Langelaar M, Threfall J, Scheutz F, Giessen JVD, Kruse H. Food-borne diseases - the challenges of 20 years ago still persist while new ones continue to emerge. Int J Food Microbiol. 2010;139:S3–S15.
Xiong Q, Cui X, Saini JK, Liu D, Shan S, Jin Y, Lai W. Development of an immunomagnetic separation method for efficient enrichment of Escherichia coli O157:H7. Food Control. 2014;37:41–5.
Yang L. Dielectrophoresis assisted immuno-capture and detection of foodborne pathogenic bacteria in biochips. Talanta. 2009;80:551–8.
Conceicao RCS, Moreira AN, Ramos RJ, Goularte FL, Carvalhal JB, Aleixo JAG. Detection of Salmonella sp. in chicken cuts using Immunomagnetic separation. Braz J Microbiol. 2008;39:173–7.
ISO 21567:2004. Microbiology of food and animal feeding stuffs -- horizontal method for the detection of Shigella spp. (http://www.iso.org/iso/catalogue_detail.htm?csnumber=34612).
ISO/DIS 6579-1: 2002. Microbiology of the food chain -- horizontal method for the detection, enumeration and serotyping of salmonella -- part 1: horizontal method for the detection of salmonella spp. (http://www.iso.org/iso/home/store/catalogue_tc/catalogue_detail.htm?csnumber=56712).
ISO/TS 21872-1:2007. Microbiology of food and animal feeding stuffs -- horizontal method for the detection of potentially enteropathogenic Vibrio spp. -- part 1: detection of Vibrio parahaemolyticus and Vibrio cholerae. (http://www.iso.org/iso/catalogue_detail.htm?csnumber=38278).
Fakruddin M, Islam S, Ahmed MM, Chowdhury A, Hoque MM. Development of multiplex PCR (polymerase chain reaction) method for detection of Salmonella spp. and Vibrio parahaemolyticus from shrimp samples of Bangladesh. Asian J Biol Sci. 2012;5(2):76–85.
Fakruddin M, Sultana M, Ahmed MM, Chowdhury A, Choudhury N. Multiplex PCR (polymerase chain reaction) assay for detection of E. coli O157:H7, Salmonella spp., Vibrio cholerae and Vibrio parahaemolyticus in spiked shrimps (Penaeus monodon). Pak J Biol Sci. 2013;16(6):267–74.
Faruque SM, Khan R, Kamruzzaman M, Yamasaki S, Ahmad QS, Azim T, Nair GB, Takeda Y, Sack DA. Isolation of Shigella dysenteriae type 1 and S. flexneri strains from surface waters in Bangladesh: comparative molecular analysis of environmental Shigella isolates versus clinical strains. Appl Environ Microbiol. 2002;68(8):3908–13.
We acknowledge the help of Mr. Mizanur Rahaman, Research Fellow, Industrial Microbiology Laboratory, IFST, BCSIR in conducting this research.
Availability of data and materials
All data and information of materials used are available in the manuscript.
Ethics approval and consent to participate
The authors declare that they have no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Fakruddin, M., Hossain, M.N. & Ahmed, M.M. Evaluation of microplate immunocapture method for detection of Vibrio cholerae, Salmonella Typhi and Shigella flexneri from food. BMC Microbiol 17, 189 (2017). https://doi.org/10.1186/s12866-017-1099-y