Determination of PCR efficiency in chelex-100 purified clinical samples and comparison of real-time quantitative PCR and conventional PCR for detection of Chlamydia pneumoniae
© Mygind et al; licensee BioMed Central Ltd. 2002
Received: 8 May 2002
Accepted: 9 July 2002
Published: 9 July 2002
Chlamydia pneumoniae infection has been detected by serological methods, but PCR is gaining more interest. A number of different PCR assays have been developed and some are used in combination with serology for diagnosis. Real-time PCR could be an attractive new PCR method; therefore it must be evaluated and compared to conventional PCR methods.
We compared the performance of a newly developed real-time PCR with a conventional PCR method for detection of C. pneumoniae. The PCR methods were tested on reference samples containing C. pneumoniae DNA and on 136 nasopharyngeal samples from patients with a chronic cough. We found the same detection limit for the two methods and that clinical performance was equal for the real-time PCR and for the conventional PCR method, although only three samples tested positive. To investigate whether the low prevalence of C. pneumoniae among patients with a chronic cough was caused by suboptimal PCR efficiency in the samples, PCR efficiency was determined based on the real-time PCR. Seventeen of twenty randomly selected clinical samples had a similar PCR efficiency to samples containing pure genomic C. pneumoniae DNA.
These results indicate that the performance of real-time PCR is comparable to that of conventional PCR, but that needs to be confirmed further. Real-time PCR can be used to investigate the PCR efficiency which gives a rough estimate of how well the real-time PCR assay works in a specific sample type. Suboptimal PCR efficiency of PCR is not a likely explanation for the low positivity rate of C. pneumoniae in patients with a chronic cough.
C. pneumoniae causes upper respiratory tract infections, and in some studies it accounts for 6–10% of community-acquired pneumonia . Seroprevalence among adults is 40–70%, increasing with age, indicating that most people are exposed at least once and that reinfections are common . C. pneumoniae is the microorganism that most commonly is associated with the inflammation seen in atherosclerosis .
C. pneumoniae infection has been detected by serological methods but PCR is currently viewed as an advantageous alternative since it detects the presence of the DNA of the organism. This allows for an early and clinically relevant diagnosis in contrast to the detection of C. pneumoniae specific antibodies that develop late in the course of the infection. Today, there is no accepted gold standard for PCR detection of C. pneumoniae, but as a start, guidelines for standardising C. pneumoniae assays have been published . Most laboratories use in-house PCR assays and DNA purification procedures. Real-time quantitative PCR in the Lightcycler is a new option among the many PCR assays and it may become a method of choice as it is fast and provides a quantitative measure. This is advantageous in a clinical setting because it allows fast diagnosis and thus fast treatment with relevant antibiotics. Furthermore, the quantitative measure could possibly be used to assess the response to treatment or to assess the state of infection; active infection versus chronic infection.
The lack of standardisation of PCR methods has been investigated in three recent studies [5–7]. These studies found strikingly different results when comparing several PCR methods. The inconsistent results were explained by the use of different DNA extraction methods, different thermal cyclers and differences in number of and/or homogeneity of replicates. Contradicting C. pneumoniae PCR results have also been explained by the fact that C. pneumoniae is often present in very low numbers because the infections tend to be chronic and low-grade . Even though most assays have very good detection limits the fact that they are often used at DNA concentrations near their detection limit can pose problems. At low DNA concentrations detection/non-detection is more susceptible to inhibitors because inhibition adds further to the statistical uncertainty already present at the detection limit (i.e. it is only possible to detect 1 copy of DNA in a fraction of experiments, because the particle distribution is expected to follow Poisson statistics). Therefore, it is necessary to test PCR methods for their performance in different situations and sample types.
The aim of this study was to compare a newly developed real-time quantitative PCR in the Lightcycler for detection of C. pneumoniae with an established conventional PCR assay used in another laboratory [10, 11] to see whether results obtained with the two methods correlate. In the real-time PCR the pmp4 gene is target and in the conventional PCR it is the 16S rRNA gene. PCR's were performed in the laboratories where they were developed. The limit of detection and performance on clinical specimens were compared and advantages and disadvantages of the methods are discussed. PCR efficiency was determined by the real-time PCR method in a random selection of the clinical samples to test whether inhibitors were present.
Results and discussion
Reference samples: comparison of the two PCR methods
Sample volumes used in the two PCR methods
Reference samples analysed with the two PCR assays. * The unit of the numbers is copies/μl and the numbers are the mean of two measurements from two different runs. # The dilution series is serial and four-fold. Dilution no. 1 has the highest concentration and dilution no. 10 has the lowest concentration. Method 1 is the real-time PCR and method 2 is the conventional PCR.
Dilution no. #
Determination of PCR efficiency in the reference and standard samples with real-time PCR
Clinical samples: comparison of the two PCR methods
Positive samples found among 136 clinical samples analysed with PCR methods 1 and 2. *copies/μl is the mean of two measurements from two different runs.
Patient sample no.
Method 1 (copies/μl)*
Spiked clinical samples: determination of PCR efficiency by real-time PCR
The efficiency of real-time PCR in the positive clinical sample with the highest C. pneumoniae DNA concentration (no. 133) was determined as described above and compared to the efficiency for the standard samples (fig. 1). A dilution series was prepared of sample 133 and the threshold cycles determined for duplicates (dilutions: 5, 10, 15, 20, 100 and 1000 times). The slope of the line (fig. 1) was -3.237 (standard error = 0.158) indicating an efficiency of 2.0. The slope for clinical sample line was not significantly different from the slope for the standard samples line (slope for the standard: -3.144 and standard error: 0.065, t-test: z = 0.54, P = 0.59).
Determination of PCR efficiency in 20 spiked clinical samples. S.E.: standard error, Eff.: Efficiency, std. standard samples. For each of 20 randomly selected negative samples spiked with C. pneumoniae DNA, the slope was determined as described and the efficiency calculated from the slope. The slope of each sample was compared to the slope of the standard from the same run with a t-test and the P-values are shown. Therefore, some of the samples are compared to the same standard.
For the remaining 17 samples the slopes were not significantly different indicating that the efficiencies should match satisfyingly between standard samples and spiked clinical samples. Hence, the PCR conditions seem to be similar in the clinical and standard samples, therefore detection and quantification is possible. Consequently, it is probably unlikely that larger numbers of positive samples were missed because of suboptimal PCR efficiency. If PCR efficiency is in part influenced by inhibitors, our results indicate that Chelex release of DNA from nasopharyngeal aspirates is sufficient in most samples. However, if the nasopharyngeal aspirates are contaminated with e.g. blood, which contain many PCR inhibitors, further purification might be needed.
It should be noted that even though slopes are not significantly different, there might still be differences in efficiencies, as the t-test allows for some variation. In table 4 it is also seen that some of the efficiencies are larger than 2, which should not be possible according to PCR theory. Some of the samples have efficiencies very much larger than 2 (samples 131 and 160) and these were also found to differ significantly from efficiencies of the standards. Substances in the samples that interfere with the PCR probably cause this. In the samples with efficiency just slightly higher than 2, it is probably caused by the fact that the slope is determined by linear regression. The linear regression does not take into account that the slope can maximally be -3.32, which corresponds to an efficiency of 2.0. Furthermore, it is notable that slopes for the standard samples vary from -3.266 to -3.716 (E varies from 1.86–2.01). Many things may cause this variation. First of all, variations in preparation of the standard dilution series is difficult to avoid because of pipetting variations, but also because DNA molecules do not behave like smaller molecules in solution. DNA is a large molecule and it is therefore likely to make intramolecular (electrostatic) interactions and also interactions with other molecules. These interactions are hard to predict as they depend on 1) which molecules are present and their concentration in the solution, 2) temperature 3) length and integrity of the DNA molecules. These phenomena probably contribute to the variation. This emphasizes that when a quantitative PCR result is presented it should be interpreted with care and always in the context of the experimental setup. When comparing quantitative measures determined in this way, it should probably be recommended to run the samples in the same run, to run at least two replicates to obtain a mean value or to include internal amplification controls. Even then the results should be interpreted with care. It is possible to detect relatively small differences between sample concentrations with quantitative real-time PCR, but the exact factor of difference between samples should be interpreted very cautiously. We have observed this previously  with our standard dilution series; in most runs it is actually possible to distinguish between e.g. 10 copies/μl and 5 copies/μl standard samples, but the factor of difference, if exact quantitative measures are compared, can vary substantially. Furthermore, the fact that the crossing points determined for the unknown samples are proportional to the log of the concentration and not directly to the concentration is also a major error factor because invlog enlarges the errors when concentration is calculated.
In practice it would not be feasible to test the PCR efficiency for every sample and in every run, because of the added time and costs. Therefore, it should be done when setting up a new real-time PCR assay and DNA purification method for a new clinical sample type. In our opinion, based on this study, the efficiency can provide a rough estimate for how well the PCR assay performs in a specific sample type, but small differences in efficiency should be interpreted cautiously because as described above many error factors are involved in the determination of this parameter.
Analysing replicates could be relevant for nasopharyngeal aspirates as C. pneumoniae might be present in very low numbers. The reason could be that cells in the nasopharyngeal aspirates stem from the superficial epithelium and C. pneumoniae might be present in larger numbers in deeper cell layers . The disadvantages of analysing replicate samples are that it adds extra costs, time and strict control of contamination. As C. pneumoniae tends to be present in low numbers especially in non-acute patients as in this study, it might be more feasible to attempt concentrating samples before or after DNA extraction [7, 14] or to improve sampling methods.
In this study sensitivity and clinical performance seems to be the same for conventional PCR and real-time PCR, but this needs to be confirmed by analysis of more clinical samples. We extended the investigation of clinical performance with determination of the PCR efficiency in the samples. The efficiencies determined by real-time PCR in clinical samples were in 17/20 cases determined not to be significantly different from efficiencies of the standards. This means that suboptimal PCR efficiency was probably not the explanation for the low positivity rate of C. pneumoniae in the nasopharyngeal aspirates. The efficiency may be used as a rough estimate for how well real-time PCR works in a given sample type.
Materials and methods
Real-time quantitative PCR was performed in a Lightcycler instrument as described previously . Briefly, the real-time PCR amplifies a 140 bp PCR product from the pmp4 gene of C. pneumoniae. The PCR fragment is detected with a fluorescent probe set using fluorescence resonance energy transfer. The pmp4 gene encodes a membrane protein from the polymorphic membrane protein (Pmp) family, which contain 21 proteins . Each gene is present in one copy on the genome, homologs are found in C. psittaci and C. trachomatis, but the sequences are not very similar on DNA level, therefore it is possible to design species specific primers and probes. Specificity was assessed by database searches with the primer and probe sequences and pmp4 gene sequences were compared for the known genome sequences of C. pneumoniae. Primer and probe sequences were found to be specific for C. pneumoniae and the pmp4 gene sequences for the known isolates were identical . Real-time PCR was performed in 20 μl consisting of: 0.5 μm of each primer, 0.2 μm of each probe, 5 mM MgCl2, 2.0 μl Faststart DNA Master Hybridisation Probes (Roche) and 2 or 6.8 μl template. The mixture was loaded into glass capillary tubes and cycling performed as previously described  The PCR was performed with 6.8 μl template DNA for analysis of the reference samples and 2 μl for analysis of the clinical samples and standard samples (see below for definition of samples). For easy comparison sample volumes are shown in table 1.
C. pneumoniae EB was purified by gradient centrifugation as previously described  and genomic DNA was purified with the Genomic-tip system from Qiagen following instructions from the manufacturer. DNA concentration and purity were analysed by agarose gel electrophoresis and measurement of OD at 260 and 280 nm. A ten-fold standard dilution series of purified C. pneumoniae genomic DNA with known concentration was used for quantification, concentrations ranged between 105 and 1 copies/μl . The Lightcycler software determined the threshold cycles for the standard samples and a standard curve was generated. The threshold cycle for unknown samples was determined for two replicates and concentrations were calculated from the standard curve also using the Lightcycler software. A negative control containing all reagents except template DNA was included in all runs. A sample was considered to be positive when the fluorescence curve was sigmoidal and reached fluorescence levels above the fluorescence threshold set in the Ligthcycler software during analysis.
A multiplex PCR with three primer sets was used which amplify fragments from the 16S rRNA gene of the three Chlamydia species. The primer pair specific for C. trachomatis, which amplifies a 240 bp product, was designed by Pollard et al. , from these primers analogues were designed for the other Chlamydia species . An internal amplification control was included in order to detect inhibition or suboptimal reaction conditions [11, 17, 18]. Amplicons were visualised with ethidium bromide and agarose gel electrophoresis. Positive results were confirmed with a C. pneumoniae specific PCR with primers detecting a 463 bp fragment of the 16S rDNA . For analysis of the reference samples 10 μl template DNA was used. For analysis of the clinical samples 10 μl was used. If inhibition occurred, the assay was repeated with 5 μl or 2 μl of the sample. Samples volumes are shown in table 1 for both PCR methods.
For both PCR methods precautions were taken to avoid contamination. Filter pipette tips were used and reagents were mixed in rooms separated from rooms where C. pneumoniae DNA from culturing, DNA purification etc. was present.
C. pneumoniae was cultivated in HeLa cells as previously described . Cells were scraped off with a rubber policeman, centrifuged and resuspended in TE-buffer. Proteinase K was added and the mixture was incubated for one hour at 55°C. Proteinase K was inactivated by incubation at 95°C for 10 min. A serial four-fold dilution series was prepared in TE-buffer with 10 dilutions ensuring a wide range of concentrations. Samples with C. pneumoniae DNA concentrations near the detection limit should therefore be present for the methods tested. In addition, two negative controls containing TE-buffer only were included. The two PCR methods were tested on the reference samples and their relative sensitivities were obtained.
Clinical samples and spiked clinical samples for efficiency determination
Nasopharyngeal aspirates were previously obtained from patients with chronic cough from the Dept. of Pulmonary Medicine at Aarhus University Hospital, Denmark. Inclusion criteria were age >15 yrs., a cough period of 2–12 weeks, a normal chest radiograph, normal spirometry and absence of chronic cardiopulmonary disease . For this study a selection of 136 samples was used. DNA was released from the nasopharyngeal aspirates by vortexing with 20% w/v Chelex 100 slurry followed by boiling of the suspensions for 10 min; then centrifuged at 20,000 × g for 10 min. . Chelex-100 is an ion-exchange resin that scavenges multivalent metal ions. PCR methods 1 and 2 were tested on these samples.
Spiked clinical samples
The real-time PCR was used for determination of PCR efficiency in a selection of clinical samples in the following way. For negative samples and samples with low C. pneumoniae DNA concentration spiking with 103 copies/μl C. pneumoniae was done. Afterwards a 2-fold dilution series with 8 dilutions was done in water with 10 μg/ml yeast RNA as carrier nucleic acid . For each of the dilutions we measured threshold cycles and plotted them towards log to relative concentrations of the dilutions. The slope was determined and compared to the slope of the standard samples from the same run.
The Danish Medical Research Council supported the research (grants no. 9900750 and 9700659). We thank Inger Andersen, Karin Skovgaard Sørensen, Birthe Dohn and Jonna Guldberg for technical assistance and Lisbet Wellejus Pedersen for linguistic assistance
- Grayston JT: Infections caused by Chlamydia pneumoniae strain TWAR. Clin Infect Dis. 1992, 15 (5): 757-761.View ArticlePubMed
- Peeling RW, Brunham RC: Chlamydiae as pathogens: new species and new issues. Emerg Infect Dis. 1996, 2 (4): 307-319.PubMed CentralView ArticlePubMed
- Saikku P: Chlamydia pneumoniae in atherosclerosis. J Intern Med. 2000, 247 (3): 391-396. 10.1046/j.1365-2796.2000.00659.x.View ArticlePubMed
- Dowell SF, Peeling RW, Boman J, Carlone GM, Fields BS, Guarner J, Hammerschlag MR, Jackson LA, Kuo CC, Maass M, Messmer TO, Talkington DF, Tondella ML, Zaki SR: C. pneumoniae Workshop Participants. Standardizing Chlamydia pneumoniae assays: recommendations from the Centers for Disease Control and Prevention (USA) and the Laboratory Centre for Disease Control (Canada). Clin Infect Dis. 2001, 33 (4): 492-503. 10.1086/322632.View ArticlePubMed
- Mahony JB, Chong S, Coombes BK, Smieja M, Petrich A: Analytical sensitivity, reproducibility of results, and clinical performance of five PCR assays for detecting Chlamydia pneumoniae DNA in peripheral blood mononuclear cells. J Clin Microbiol. 2000, 38 (7): 2622-2627.PubMed CentralPubMed
- Apfalter P, Blasi F, Boman J, Gaydos CA, Kundi M, Maass M, Makristathis A, Meijer A, Nadrchal R, Persson K, Rotter ML, Tong CY, Stanek G, Hirschlm AM: Multicenter Comparison Trial of DNA Extraction Methods and PCR Assays for Detection of Chlamydia pneumoniae in Endarterectomy Specimens. J Clin Microbiol. 2001, 39 (2): 519-524. 10.1128/JCM.39.2.519-524.2001.PubMed CentralView ArticlePubMed
- Smieja M, Mahony JB, Goldsmith CH, Chong S, Petrich A, Chernesky M: Replicate PCR Testing and Probit Analysis for Detection and Quantitation of Chlamydia pneumoniae in Clinical Specimens. J Clin Microbiol. 2001, 39 (5): 1796-1801. 10.1128/JCM.39.5.1796-1801.2001.PubMed CentralView ArticlePubMed
- Boman J, Gaydos CA, Quinn TC: Molecular diagnosis of Chlamydia pneumoniae infection. J Clin Microbiol. 1999, 37 (12): 3791-3799.PubMed CentralPubMed
- Mygind T, Birkelund S, Falk E, Christiansen G: Evaluation of real-time quantitative PCR for identification and quantification of Chlamydia pneumoniae by comparison with immunohistochemistry. J Microbiol Meth. 2001, 46: 241-251. 10.1016/S0167-7012(01)00282-2.View Article
- Birkebæk NH, Jensen JS, Seefeldt T, Degn J, Huniche B, Andersen PL, Østergaard L: Chlamydia pneumoniae infection in adults with chronic cough compared with healthy blood donors. Eur Respir J. 2000, 16 (1): 108-111. 10.1034/j.1399-3003.2000.16a19.x.View ArticlePubMed
- Storgaard M, Østergaard L, Jensen JS, Farholt S, Larsen K, Ovesen T, Nødgaard H, Andersen PL: Chlamydia pneumoniae in children with otitis media. Clin Infect Dis. 1997, 25 (5): 1090-1093.View ArticlePubMed
- Belotserkovskii BP, Johnston BH: Denaturation and association of DNA sequences by certain polypropylene surfaces. Anal Biochem. 1997, 251 (2): 251-262. 10.1006/abio.1997.2249.View ArticlePubMed
- Kauppinen M, Saikku P: Pneumonia due to Chlamydia pneumoniae: prevalence, clinical features, diagnosis, and treatment. Clin Infect Dis. 1995, 21 (Suppl. 3): S244-S252.View ArticlePubMed
- Maass M, Jahn J, Gieffers J, Dalhoff K, Katus HA, Solbach W: Detection of Chlamydia pneumoniae within peripheral blood monocytes of patients with unstable angina or myocardial infarction. J Infect Dis. 2000, 181 (Suppl 3): S449-S451. 10.1086/315610.View ArticlePubMed
- Kalman S, Mitchell W, Marathe R, Lammel C, Fan J, Hyman RW, Olinger L, Grimwood J, Davis RW, Stephens RS: Comparative genomes of Chlamydia pneumoniae and C. trachomatis. Nat Genet. 1999, 21 (4): 385-389. 10.1038/7716.View ArticlePubMed
- Knudsen K, Madsen AS, Mygind P, Christiansen G, Birkelund S: Identification of two novel genes encoding 97- to 99-kilodalton outer membrane proteins of Chlamydia pneumoniae. Infect Immun. 1999, 67 (1): 375-383.PubMed CentralPubMed
- Pollard DR, Tyler SD, Ng CW, Rozee KR: A polymerase chain reaction (PCR) protocol for the specific detection of Chlamydia spp. Mol Cell Probes. 1989, 3 (4): 383-389.View ArticlePubMed
- Tarp B, Jensen JS, Østergaard L, Andersen PL: Search for agents causing atypical pneumonia in HIV-positive patients by inhibitor-controlled PCR assays. Eur Respir J. 1999, 13 (1): 175-179. 10.1034/j.1399-3003.1999.13a32.x.View ArticlePubMed
- Gaydos CA, Quinn TC, Eiden JJ: Identification of Chlamydia pneumoniae by DNA amplification of the 16S rRNA gene. J Clin Microbiol. 1992, 30 (4): 796-800.PubMed CentralPubMed
- Clausen JD, Christiansen G, Holst HU, Birkelund S: Chlamydia trachomatis utilizes the host cell microtubule network during early events of infection. Mol Microbiol. 1997, 25 (3): 441-449. 10.1046/j.1365-2958.1997.4591832.x.View ArticlePubMed
This article is published under license to BioMed Central Ltd. This is an Open Access article: verbatim copying and redistribution of this article are permitted in all media for any purpose, provided this notice is preserved along with the article's original URL.